Looking for portable nebulizer? Here is Compact Ultrasonic Nebulizer for home use, it is so compact that you can carry with yourself while travelling. It is very easy to use, fill warm water in lower cup, add medicine in upper cup, fit the top cap and plug appropriate extension for your medical condition and switch it on. As simple as that and portable to carry with you anywhere in emergency condition. It comes with different mouth piece, one for adult and one for child and extension as well. Different mouth pieces are used for different medical problem like cough, asthma and bronchitis.
Two airflow control,
Compact and portable,
Suitable for all ages people.
1. HKz : 2.4MHz
2. Particle size : Res pirable Fraction 0.5
3. Maximum Med capacity : 15ml
4. Machine Capacity : 25ml
5. Average Nebulization Rate : 0.4ml/min
6. Working voltage : 12V DC
7. Power Consumption : 6w
8. Weight : 0.5KG (Approx.)
9. Input voltage: 110V-240V
Usage Instructions :
2. Add 30ml purified water into the Water Chamber, the water level should be 3-5mm above the external silicon ring.
3. Fill the medicine cup half-full. put the cup into the chamber.
4. Re-install the top cover securely with the silicon gasket.
(Note: the best atomisation temperature in the chamber is 30℃/8℉)
Please use saline water or WARM(NOT boiling) water to sterilize the accessories.
- The better way use about 20-60 degrees hot water pour into big cup when you first time use it, only can be use clear water.
- Please fill purified and hot water in water cup under medicine cup before starting machine, otherwise it will damage the machine.
- Please add purified water instead of running water into the water chamber.
- The machine should be cleaned promptly after use.
- Please check whether the machine is damaged, cracks, est. before and after using the product. If yes, please contact our customer service for replacements.
- The machine itself does not have the heating function.
- If the machine doesn’t work or red light flashes, check whether there’s enough water in the water chamber, ensure that the Atomizing Cup can touch the water surface.
- If completely no mist or small mist, check the following:</li
- If the water temperature is below 20℃/68℉, replace it with water at 40℃/104℉.
- If the medicine liquid has a high concentration or small quantity, it needed to be diluted with normal saline before operate.
- If the liquid for automation contains large particles/impurity/precipitat/flatage/colloidal particles, the machine would not work.
- Change a NEW cup, if no mist after few uses.
- If the top cover does not fit, please ensure the silicon gasket been placed tightly.
- For water leakage issue, please don’t make the machine tilt.
Package Includes :
1 x Portable Nebulizer
2 x Mask (1 adult size, 1 children size)
5 x Medical cups
2 x Mouth pieces
1 x AC adapter
1 x Manual